Tehn Can you please help me what to do. Using PEEK frameworks for full arch implant supported prosthetics. The Kodak trademark and trade dress are used under license from Kodak. Faustinio I am also having KV error and the system beeping. Excuse me here is Amir Tabatabaei again. Back to news USA.
|Published (Last):||24 August 2005|
|PDF File Size:||8.74 Mb|
|ePub File Size:||15.96 Mb|
|Price:||Free* [*Free Regsitration Required]|
Page Using The System Using the System Every dental specialist would like to produce high-quality intraoral radiographs that reveal maximum detail with the minimum dose to the patient, show teeth and anatomic structures accurately with a minimum of distortion or magnification, and have optimal density and contrast to maximize their use for the detection of dental diseases.
Page Exposure Exposure Exposure Parameters Because each receptor film, digital sensor, or phosphor plate has its own sensitivity to x-ray radiation, the choice of receptor affects the exposure parameters.
For instance, sensitivity class for conventional dental films is characterized with a letter D, E, or F where F is more sensitive than E, and E more sensitive than D. Page Processing Note 3. Make an exposure. Exposure times range from 0. Press the radiology control key. Page Cleaning Cleaning Clean the outside of the system with a damp paper towel or soft cloth using an alcohol-based, non-corrosive cleaner.
Disinfecting If necessary, wipe off surfaces with disinfectant. Page Troubleshooting Troubleshooting Table Connect the unit. Fuse F1 is burned out or defective. Replace the fuse.
Circuit breaker is off. Turn on the circuit breaker. Control unit does not Control unit is disconnected. Connect the control unit. Classification: Group 1, Class B. Page X-ray Generator Filtration due to fixed materials 0. Al Cooling Figure Equipped x-ray generator IEC standard Conform Type of protection against electric shocks Class I Degree of protection against electric shocks Type B Rated value of inherent filtration 1.
Al Rated value of additional filtration 1. Page 31 Cooling Time in minutes Figure
Kodak 2200 User Manual
KODAK 2200 INTRAORAL X-RAY SYSTEM PDF
Carestream CS 2200 Intraoral X-ray System
Class 2 Device Recall Model Kodak 2100 Intraoral XRay System